GP SPHEROID RBA
SPHEROID project began in Athens.
test SPHEROID GP beta 01 2.2Resistance 015Wire 3.7Volt on Vimeo
test SPHEROID GP beta 01 1.9Resistance 017Wire 3.8Volt on Vimeo
The form of the prototype Spheroid, our first atomizer below.
Spheroid Beta01 had brass 510 connection, it was ok for a week, but finally the 510 connection broke, then followed the beta02 full SS which became unbreakable, new internal shape changes for better taste, we are finally satisfied with its throat-hit and taste. Now beta03 the last version, is in the CNC for some modifications to its external shape and a few small details in general.
Photo of Beta01 with brass 510 connection, now it is SS.
Spheroid v1 is now complete after 4 months of testing, 6 months after its creation and 5 internal changes.
GP Spheroid SS v1 on Vimeo
The direction of the airflow, the diameter and the position of the airflow, the small chamber, the quantity of the inhaling vapour, etc were the parameters that occupied us in achieving a wick type aty with one wire, without ceramic, average capacity, one having rich taste with strong throat hit, small dimensions and simple setup. We believe we did it. Its feeding system works so well that you won't take a burned taste even with a very low resistance, if the set up has been done properly. Its suction system gives a signal of weak taste for about 10 draws, before it runs out completely of liquid.
The vapour production and the throat hit depends on how low or high you have put the resistance wire. Low position gives more vapour and strong throw heat. High position gives less vapour and extreme throat hit.
GP Spheroid v2
Updated materials (SS 316, PEEK plastic insulators), increased machining precision (CNC with latest technology tools like electrodynamic drilling, better threading) and a new logo.
GP Spheroid Animation
We are developing the GP Spheroid v3, no ETA yet.